You might be wondering where you can buy a compatible GE Logiq C5 probe
. The answer is surprisingly simple. All you need is a compatible transducer. Here are a few places that sell compatible GE probes. Just make sure that you purchase the compatible one and keep in mind that it may not be exactly the same. Read on to find out more about your options.
GE 7S-RS transducer
A GE Logiq S7 ultrasound machine produces top-notch results with its 3D imaging technology and advanced features. This unit’s ergonomic design and versatility makes it ideal for a wide range of clinical applications. Its high-definition imaging system offers enhanced diagnostic capabilities, including volume enhancements and a B-Flow display. There are several specialty probes to fit any GE ultrasound machine.
In the BT15 version, the transducer is equipped with an integrated transmitter and receiver board, or Voluson E8. If the Voluson E8 is faulty, the resulting image will be abnormal. Otherwise, the APS power supply, RFM, and front and rear connections are all defective. Hence, a replacement is necessary.
A GE 7S-RS transducer is compatible with a GE Logiq C5 probe, which features a patented wide-band technology. It transmits ultrasonic pulse signals without gaps, and the signal is amplified after reception. Digital coding prevents distorted signals and increases the dynamic range. It is recommended for use in pediatrics and gynecology.
The NAIM analog voltage modulation board is responsible for the front-end 320MHz master clock and provides pulse voltage for exciting the probe. The NAIM board also monitors front-end temperature and manages power supply. Heat dissipation could lead to a damaged NAIM board. R650 resistor controls the high-voltage dynamic change part.
The GE Logiq e mainframe shuts down prematurely after only half-starting. This error can be caused by several different factors, but the most common causes are listed below:
A TI board can also fail. A failure of this board can cause a 2D area with no image, a card screen, and a boot code stating “US_IMG_17”. In addition to this, the PSA is responsible for the front-end power supply. If the PSA fails, the probe will no longer function. A GE 7S-RS transducer is compatible with GE Logiq C5 probe
GE i12L-RS transducer
The i12L-RS linear array transducer from GE is a highly versatile linear transducer with a frequency range of 5.0 to 13.0 MHz. It is suitable for peripheral vascular, musculoskeletal, and small parts applications. The i12L-RS transducer is compatible with the GE Voluson e, i, and S6 ultrasound probes.
It is compatible with a wide range of departments and has advanced imaging technologies including Speckle Reduction Imaging (CHI), CrossXBeam and Tissue Harmonics. This imaging system is equipped with 3 transducer ports and offers comprehensive patient data. GE Logiq P6 ultrasound systems are easy to use and very affordable. The system is equipped with an intuitive keyboard and features all the required keys around a trackball.
GE ultrasound machines and scanners are compatible with GE’s newest generation of transducers and probes. With over 100 different types to choose from, WSA electronic Ukraine, Ltd. offers a variety of high-quality ultrasound probes for a range of applications. Some of the common types of ultrasound probes include linear, convex, and phased array. There are also special probes for specific applications such as cardiovascular, general imaging, and women’s health.
One common problem with the GE 7S-RS compatible GE LOGIQ C5 color Doppler ultrasound probe is that the probe frequently restarts when it first boots. This is often caused by a malfunction in the computer, which serves as the back end of the equipment. The computer may be damaged, or the system board is loose. To fix the problem, you will need to remove the software and rebuild the system from scratch.
The power supply for the ultrasonic probe is not recognized. Normally, the probe interface board is powered by the machine’s power supply. On a machine’s configuration diagram, there is no high-voltage switch. However, this does not affect the high-voltage output of the DC power supply. When a probe cannot recognize a power supply, the next step is to replace the transmitter board and probe.
Another common problem with a GE Logiq C5 probe is the lack of echo. The probe may not be receiving enough echo when scanned. When this occurs, the transmitter board may burn. In addition, it may result in an abnormal image. You should try to measure the probe’s impedance to ground with a multimeter. For easier testing, you can also measure its impedance to ground without using the host.
Some users report that they are unable to diagnose the problem by resetting the machine. However, many users are able to solve this issue by unplugging their probes, performing a self-check, and restoring the mirror backup software. If this problem persists, you can try a different GE 7S-RS compatible GE Logiq C5 probe.
After the replacement of the GE Logiq C5 probe, you should check the error message that appears on the machine. It appears to be due to a hardware problem. Normally, a faulty probe may cause the machine to malfunction. Nevertheless, if the problem is caused by the sensor itself, the replacement device should be able to resolve the issue. Alternatively, you can also use a USB flash drive to copy the log data. Afterward, you can compare the scanned image to a reference.
In some cases, the faulty probe may cause the GE 7S-RS compatible GE LOGIQ e mainframe to shut down after only half startup. In such a case, you may need to replace the probe and the mainboard battery. If you do not want to go through the hassle of replacing the probe and the mainframe, change the BIOS settings to the Philips Settings.